" Innovation for Precision "
If you want to set-up the D-Cube according to the isocenter;
1. Align the D-Cube to the cross-hair roughly at Gantry:0°
2. Align the D-Cube to the cross-hair roughly at Gantry:90°
3. Ready for Tests!
You do not need to align the D-Cube to the cross hair perfectly since D-Cube detects the exact position of cross-hair and its angle. And these values are used for reference isocenter. Therefore you can start your tests by eliminating user dependent uncertainties. We will continue to give information about unique features of D-Cube Smart QA System and UFC. Follow us #MD Research & Development.